Angle of Incidence:
0 ± 5 °
OD > 4.5 506.5 nm
ODavg > 10 516 – 556 nm (Design specification – measurements are noise-floor limited)
ODavg > 2 925 – 1100 nm
ODavg > 2.5 700 – 925 nm
ODavg > 5 200 – 400 nm
ODavg > 5 400 – 457.5 nm
ODavg > 5 556 – 700 nm
≥ 21 mm
Filter Effective Index:
1.9 Understanding 'Effective Index of Refraction' neff
Filter Thickness Tolerance, Mounted:
Filter Thickness, Mounted:
Guaranteed Minimum Bandwidth:
Mounted / Unmounted:
Optical Damage Rating:
Testing has proven to show no signs of degradation when exposed to at least 6.0 W of power from an unfiltered xenon arc lamp over a 25 mm diameter (corresponding to 1.2 W/cm2) for over 500 hrs.
Arrow on ring indicates preferred direction of propagation of light
BrightLine Single-Band Bandpass Filter
Semrock Filter Family:
BrightLine® Fluorescence Filters
21.8 x 2.0 mm
low-autofluorescence optical quality glass
Surface Quality (Scratch-Dig):
Tavg > 93% 464.5 – 499.5 nm
Transverse Tolerance, Mounted:
+ 0.0 / - 0.1 mm
Individual fluorescence bandpass filters that have been optimized for use in a variety of fluorescence instruments. All thin-film, hard-coated construction for unsurpassed performance and reliability.
Unfortunately there is too much information to see on such a small screen. Please find a large tablet, laptop or a desktop computer.
Or you can get in touch with us.
Get in touch
December 10-14, 2018
The Sauer Laboratory at the University of Würzburg implements Re-scan Confocal Microscopy with a new turnkey system from Confocal.nl.